A Holter monitor is an ambulatory electrocardiographic system discovered by Dr. Norman J. Holter and his team in 1957. Little did they knew that their device would change the way we track cardiac rhythms out of the hospital.
Historically electrocardiography began in 1893 with the work of Einthoven's string galvanometer. The Holter monitor is a device that works on Galvanometer's principle to record electrocardiographic signals from an individual who is going about his daily activities, like continuous ambulatory electrocardiography. From 1961 various other modalities and advance gadgets have been manufactured for this purpose.
Since then it has formed the backbone of rhythm detection and analysis in Cardiac Electrophysiology.
There are no relevant recommendations available in the literature to distinguish patients who may benefit from the application of ambulatory electrocardiogram monitoring. The present proposals are as follow:
The selection between the common two to three lead or the twelve lead Holter ECG monitoring depends mostly on the desired goal. If it is being used to monitor heart rate and its rhythm then two to three leads are sufficient. In contrast, if the purpose is to establish the origin of premature beats/ dysrhythmias or tachycardia, then a twelve-lead Holter electrocardiography is the preferred modality of choice.
Selection of which monitor to use depends on the frequency of symptoms. When symptoms are continuous, a routine twelve-lead electrocardiogram is sufficient to make a diagnosis. Usually, Cardiologists use the Holter monitor for intermittent symptoms. If symptoms rarely occur, the devices for the longer duration like Implantable loop recorder (ILR) or event monitor can be used.
A twelve-lead Holter monitor is very accurate and can instantly diagnose supraventricular tachycardia (SVT), ventricular tachycardia (VT), atrial flutter, atrial fibrillation, monomorphic or Polymorphic VTs, long QT syndrome, supraventricular premature complexes, ventricular premature complexes, dominant atrioventricular accessory pathways, atrioventricular block, right and left bundle branch block, and left anterior and posterior fascicular block.
Current European guidelines endorse the use of an intracardiac monitor for patients diagnosed with a cryptogenic stroke.
Mobile electrocardiographic monitoring is contraindicated if it delays urgent treatment, hospitalization or a procedure. For example, it should not be part of the initial investigation for angina, where a stress test would be more appropriate.
The monitor is also contraindicated in patients who have syncope and high-risk factors at which time inpatient management is mandatory.
Patients with symptoms such as syncope, near-syncope, episodic dizziness, or palpitation in whom other clear causes have been identified by history, physical examination, or laboratory tests.
The ACC/AHA guidelines discouraged the use of ambulatory ECG for either arrhythmia detection or analysis of heart rhythm variability for risk assessment in patients without symptoms of arrhythmia, even if they had cardiovascular conditions such as left ventricular hypertrophy, or valvular heart disease.
A patient who refuses to undergo further therapy once arrhythmia is established. Routine screening of asymptomatic patients.
The initial Holter monitor was about the size of a briefcase and comprised of an amplifier, tape recorder, electrodes attached to the individual's chest, playback unit and analyzing unit. It consisted of ten electrocardiograms leads including six standard precordial electrodes and four torso electrodes. The function of torso leads is to prevent signal disturbances. However, torso leads can show variations. To overcome these variations, investigators proposed mathematical algorithms. While initially consisting of 12 leads recent models have comprised of as little as two to three leads.
After continuous improvement and progress, the Holter is now the size of a small cell phone and gives two types of primary data to analyze. One is the QRS complex, and the other is the R-R interval. It continuously records until it is detached from the patient or it runs out of power, although it is usually used for 24-48Hrs. The power supply lasts 80-100 hours with a tape recording capacity of ten hours.
The Physician will explain the procedure and should educate the patient about how to record the event if it happens. A physician should inform patients regarding serious events like chest pain, shortness of breath or lightheadedness. The device should be interrogated within 24 hours of the event to prevent further event and to take necessary actions if needed.
Unlike invasive procedures, no preparation is needed, but the patient should be advised to keep the monitor away from other electrical devices while wearing the device. Physician should recommend not to put lotion or moisturizer on the chest as it will affect the attachment of leads.
Holter monitoring continuously records an Electrocardiogram (EKG) tracing on three channels for 12–48 hours. The patient activates a button to correlate the timing of symptoms with the EKG. Cardiac technician and skilled physician examine entire recorded EKG for abnormalities
In children, Holter monitor use is limited because of wires and cables. In 2011, the Federal Drug Administration approved a device which is wireless, water-resistant and allows for up to 14 days of continuous single-channel rhythm monitoring. The patients can trigger symptoms by pressing a button to start the recording and correlate symptoms with concurrent rhythm. A study which compared new wireless device with standard Holter monitor found that new wireless adhesive monitoring patch diagnosed more arrhythmic events than the Holter monitor.,
The device stays in a pocket outside of the body close to patients chest, usually in the form of a neck sling or in the patient's vest pocket. So theoretically there aren't any complications, but the surface electrodes can irritate the underlying skin and if left in place can cause skin ulceration, but this rarely ever happens as the device technicians remove them when they take the machine off the patient.
Utilization of the Holter monitor has significantly increased, especially in the detection of occult atrial fibrillation as a cause of a cryptogenic stroke. Anticoagulation is always superior to antiplatelet in the secondary prevention of stroke due to atrial fibrillation. So, the diagnosis of occult atrial fibrillation to initiate anticoagulation with the help of the Holter monitor can prevent recurrent strokes.
Patients with LV systolic dysfunction and presumed arrhythmic event as well as those with symptoms supposed to be due to transient second or third-degree heart blocks also form a significant number of people who undergo the test.
Holter monitors are excellent resources to look for and diagnose cardiac arrhythmias in the appropriate patient populace. Clinicians and physician assistants should be aware of the possibility of arrhythmias as usually they present with non-specific symptoms, so a high index of suspicion and prompt referral/ recommendation are essential in this regard.
With the advent of highly effective pharmacotherapy and interventional treatment of cardiovascular ailments, the pool of survivors and life expectancy is ever-growing, and with this, the potential substrate for arrhythmias is also increasing daily, so Holter monitors can provide more information when appropriately utilized.
We are looking for contributors to author, edit, and peer review our vast library of review articles and multiple choice questions. In as little as 2-3 hours you can make a significant contribution to your specialty. In return for a small amount of your time, you will receive free access to all content and you will be published as an author or editor in eBooks, apps, online CME/CE activities, and an online Learning Management System for students, teachers, and program directors that allows access to review materials in over 500 specialties.
This is an academic project designed to provide inexpensive peer-reviewed Apps, eBooks, and very soon an online CME/CE system to help students identify weaknesses and improve knowledge. We would like you to consider being an author or editor. Please click here to learn more. Thank you for you for your interest, the StatPearls Publishing Editorial Team.
Click Your Answer Below
Would you like to access teaching points and more information on this topic?
|HOLTER NJ,GENERELLI JA, Remote recording of physiological data by radio. Rocky Mountain medical journal. 1949 Sep [PubMed]|
|Pevnick JM,Birkeland K,Zimmer R,Elad Y,Kedan I, Wearable technology for cardiology: An update and framework for the future. Trends in cardiovascular medicine. 2018 Feb [PubMed]|
|Giada F,Bartoletti A, Value of Ambulatory Electrocardiographic Monitoring in Syncope. Cardiology clinics. 2015 Aug [PubMed]|
|Diemberger I,Gardini B,Martignani C,Ziacchi M,Corzani A,Biffi M,Boriani G, Holter ECG for pacemaker/defibrillator carriers: what is its role in the era of remote monitoring? Heart (British Cardiac Society). 2015 Aug [PubMed]|
|Wang R,Blackburn G,Desai M,Phelan D,Gillinov L,Houghtaling P,Gillinov M, Accuracy of Wrist-Worn Heart Rate Monitors. JAMA cardiology. 2017 Jan 1 [PubMed]|
|Jakicic JM,Davis KK,Rogers RJ,King WC,Marcus MD,Helsel D,Rickman AD,Wahed AS,Belle SH, Effect of Wearable Technology Combined With a Lifestyle Intervention on Long-term Weight Loss: The IDEA Randomized Clinical Trial. JAMA. 2016 Sep 20 [PubMed]|
|Mâsse LC,Fuemmeler BF,Anderson CB,Matthews CE,Trost SG,Catellier DJ,Treuth M, Accelerometer data reduction: a comparison of four reduction algorithms on select outcome variables. Medicine and science in sports and exercise. 2005 Nov [PubMed]|
|Prabhu S,Taylor AJ,Costello BT,Kaye DM,McLellan AJA,Voskoboinik A,Sugumar H,Lockwood SM,Stokes MB,Pathik B,Nalliah CJ,Wong GR,Azzopardi SM,Gutman SJ,Lee G,Layland J,Mariani JA,Ling LH,Kalman JM,Kistler PM, Catheter Ablation Versus Medical Rate Control in Atrial Fibrillation and Systolic Dysfunction: The CAMERA-MRI Study. Journal of the American College of Cardiology. 2017 Oct 17 [PubMed]|
|Chai PR, Wearable Devices and Biosensing: Future Frontiers. Journal of medical toxicology : official journal of the American College of Medical Toxicology. 2016 Dec [PubMed]|
|George L,Gargiulo GD,Lehmann T,Hamilton TJ, Concept Design for a 1-Lead Wearable/Implantable ECG Front-End: Power Management. Sensors (Basel, Switzerland). 2015 Nov 19 [PubMed]|
|Lobodzinski SS, ECG patch monitors for assessment of cardiac rhythm abnormalities. Progress in cardiovascular diseases. 2013 Sep-Oct [PubMed]|
|Albers GW,Bernstein RA,Brachmann J,Camm J,Easton JD,Fromm P,Goto S,Granger CB,Hohnloser SH,Hylek E,Jaffer AK,Krieger DW,Passman R,Pines JM,Reed SD,Rothwell PM,Kowey PR, Heart Rhythm Monitoring Strategies for Cryptogenic Stroke: 2015 Diagnostics and Monitoring Stroke Focus Group Report. Journal of the American Heart Association. 2016 Mar 15 [PubMed]|
The intent of StatPearls is to provide practice questions and explanations to assist you in identifying and resolving knowledge deficits. These questions and explanations are not intended to be a source of the knowledge base of all of medicine, nor is it intended to be a board or certification review of Nurse-Adult Medical Surgical RN. The authors or editors do not warrant the information is complete or accurate. The reader is encouraged to verify each answer and explanation in several references. All drug indications and dosages should be verified before administration.
StatPearls offers the most comprehensive database of free multiple-choice questions with explanations and short review chapters ever developed. This system helps physicians, medical students, dentists, nurses, pharmacists, and allied health professionals identify education deficits and learn new concepts. StatPearls is not a board or certification review system for Nurse-Adult Medical Surgical RN, it is a learning system that you can use to help improve your knowledge base of medicine for life-long learning. StatPearls will help you identify your weaknesses so that when you are ready to study for a board or certification exam in Nurse-Adult Medical Surgical RN, you will already be prepared.
Our content is updated continuously through a multi-step peer review process that will help you be prepared and review for a thorough knowledge of Nurse-Adult Medical Surgical RN. When it is time for the Nurse-Adult Medical Surgical RN board and certification exam, you will already be ready. Besides online study quizzes, we also publish our peer-reviewed content in eBooks and mobile Apps. We also offer inexpensive CME/CE, so our content can be used to attain education credits while you study Nurse-Adult Medical Surgical RN.